type molten iron component analyzing device
is made of high accuracy temperature converting
tal equipment and sigle silicom. When using
digital type molten iron component analyzing
device, user only pours the original molten
iron into CE-Cup and waits for 150s and then
gets the measured results.
It can show the CE, content of carbon and silicon
clearly with high-light red LED large screen
with 3 inches per digit in 3 rows 5 lines. When
undercooling molten iron appears, digital screen
will warn the operator to discard the measured
result so as to prevent quality mislead by measured
measuring rang of digital type molten iron component
analyzing device: CE2.50-4.80、C%2.30-4.20、Si%0.60-3.80.
Measured accuracy of single grade molten iron
can be guaranteed within the range of error
||Digital type molten
iron component analyzing device has no function
of materials adjustment. When charging materials
are not stable and thus cause error, operator
can apply for the service of adjustment. Writing
calibration formula into the digital type molten
iron component analyzing device will keep the
high accuracy of measued results.
This device is simple, cheaper , easier and
reliable quality. It will satisfy to the user
who produces high volume of single product.